FluidFM User Information Center
AFM Solutions: FluidFM Add-on for AFM
ADD-ON FAQ 3 essays
- The signal noise is too large, what can I do?
- Petri-dishes for the FluidFM ADD-ON
- Download CORA software for the FluidFM ADD-ON
Cell + Bioscience: FluidFM BOT
FluidFM BOT FAQ 1 essay
FluidFM BOT Troubleshoot 1 essay
FAQ (General) 4 essays
- Why does the microfluidics controller make a click noise?
- I cannot align the laser, what now?
- How to report a problem?
- Where are the log-files?
3D Printing: FluidFM µ3Dprinter
Probes FAQ 11 essays
- What is the angle of a FluidFM probe?
- How do I fill FluidFM probes correctly?
- How long can FluidFM probes be used?
- Can I exchange the liquid in the probe?
- Correct storage for FluidFM probes
- What is the effective sensitivity of FluidFM probes?
- How to prevent a clogged probe?
- How can I avoid air bubbles in the probe?
- Why is there no droplet coming out at the end of the probe during filling?
- My probe dried out during storage. Will it still work?
- How thick are the channels and walls of FluidFM micropipettes?
Probe Troubleshoot 1 essay
Application protocols 2 essays